74HCT08N datasheet, 74HCT08N pdf, 74HCT08N data sheet, datasheet, data sheet, pdf, NXP Semiconductors, Quad 2-input AND gate. 74HCT08N datasheet, 74HCT08N circuit, 74HCT08N data sheet: PHILIPS – Quad 2-input AND gate,alldatasheet, datasheet, Datasheet search site for. 74HCT08N 74HC08; 74HCT08; Quad 2-input And Gate;; Package: SOT ( SO14), SOT (DIP14) Details, datasheet, quote on part number: 74HCT08N .
|Published (Last):||15 January 2013|
|PDF File Size:||9.77 Mb|
|ePub File Size:||4.5 Mb|
|Price:||Free* [*Free Regsitration Required]|
An error occurred the shopping list s could not be be loaded. The measured values detected by the calibration Calibration in a controlled environment measuring laboratory. This additional time for the calibration should be taken into consideration to your normal delivery time. Please enter your name and your e-mail address.
After that, I debug using ICD2 and i got 0 for variable “i”. But the closest I could get is the company that manufacture it and other info that related to that company.
74HCT08N – 74HCT Quad/2-Input/AND DIL-14 – NXP
Please, I want to get something working here, and I really need your help. I tried to find the datasheet for you. Introduction In collaboration with Intertek, Fatasheet is offering a calibration service for new electronic test and measurement instruments.
Thank you for your message.
I really don’t know what to say for the effort that you are doing with me. Xavier38 New Member Total Posts: Close An error occurred, your message could not be sent. This one is the Ethernet Socket that has a transformer. After the order has been datahseet for the calibrated version, the non-calibrated dagasheet will be sent out from our warehouse to the calibration institute of Intertek Calibration service in process at Intertek Creating the certificate for end user The device will be repacked and sent directly to the end user.
However, even that datasheet also cannot be obtained due to some problem with their server i guess. Top Brands Datashret All Manufacturers. Can I return the calibrated item if I am not satisfied with the device itself?
74HCT08N datasheet: Quad 2-input AND gate- documentation on electronic component free download
I went to a computer shop next to my home that fixes computer boards. However, after i datasheer the board and connect the board to the router, the router’s indicator light did not light up.
Those are the main pins. Usually the time for the calibration of your device can be handled within 5 working days.
Log in to see your specific prices. My router have 4 ports, and lets say, port 1 is connected to the computer, the indicator light at port 1 will light up. You can visit them or email them email address is in their website to request the datasheet directly from them. San Antonio, Texas Status: Pins 45, 46, 58, 59 respectively.
Hopefully somebody who understands the code can help me to see my mistake. As the product is customised to your requirements, we cannot accept returns should you decide you no longer want the instrument. And when I asked him about the Ethernet Socket that had a transformer, datasjeet had no clue -what so ever- about what I was talking about.
An error occurred login failed. Expert technical support on hand Our expert team have over 25 years experience solving problems and providing expert advice. Calibration involves comparing the read values of a measuring instrument or data recorder with the corresponding values of a reference tool that meets the reference standards.
An error occurred the product could not be saved. I have an ethernet with a transformer but I can’t find its datasheet at all is there a way to know its datasheeg or anything?? 74hc0t8n have reconstruct the board on breadboard, test the relevant voltages, change a new 25MHz oscillator and capacitor as well as use Silicon Rev B5 resistor.
Items 74hhct08n are on stock will be shipped out for calibration within 24 hours. Anybody use 74HCT08N as level shifter before.? You will find two items appearing, one version without a calibration service item-nr. The bolded line are the difference between 2 debugging.